Session: Sep 1999 (session 3)
Wavelength: 18 cm
To give feedback on this experiment, click
Noto (C.Trigilio at Mon Sep 27 11:37:58 METDST 1999) Success
Medicina (Sergio Mariotti at Mon Sep 27 11:56:11 METDST 1999) Success
Nothing to say
Torun (K.Borkowski at Mon Sep 27 14:34:29 METDST 1999) Success
Little more than 1 min. lost beginning at 0:07:30 due to
a long slew around the azimuth axis to a new source.
Effelsberg (A. Kraus at Tue Sep 28 10:50:16 METDST 1999) Minor failures
Some scans were missed because of insufficient slewing time. Apparently,
we had some problems with the thin tape on the VLBA4 recorder (many
Jodrell (Alastair Gunn at Wed Sep 29 13:34:08 METDST 1999) Failure
The Lovell telescope had been transferred to another program
in the break before this experiment, and the LO/IF settings
not returned to their correct values. All data therefore useless.
Calibration (Jean\-Francois Desmurs at Sun Nov 7 15:56:13 MET 1999) Minor failures
The ANTAB formatted file for this experiment can be found on VLBEER
in the sep99 directory under the name:
Read the comments in the file to get further information
(Rem : Jb bad Tsys values.)
JIVE (Bob Campbell at Wed Mar 1 08:56:57 MET 2000) Minor failures
No Jb or Tr throughout -- hence 5 stations.
Wb tracks 7,9,10,15 marginal
--> RCP 3rd subband, LCP 1st & 3rd subband could be suspect (arising from
the interplay between subband/polarization channels & their constituent
tracks via fan-in).
aips++ plots given to Co-Inv. during visit to JIVE: all weight(t),
amp(lag); selected amp(t), phs(t), amp(frq).