Sep 1997 (session 3)
To give feedback on this experiment, click
Medicina (Sergio Mariotti at Mon Sep 29 00:35:33 METDST 1997) Success
Onsala (Antonis Polatidis at Mon Sep 29 10:11:50 METDST 1997) Success
Some problems with positioning head 2, this should not affect the quality of the data.
Effelsberg (dgraham at Mon Sep 29 11:58:55 METDST 1997) Minor failures
Long drive times between sources. 3C84 scheduled below elv limit.
Noto (C.Trigilio at Mon Sep 29 14:27:07 METDST 1997) Minor failures
Many sources below horizont! THE LOWEST ELEVATION IN NOTO IS 5 DEG!
OAN\-Yebes (P. de Vicente at Mon Sep 29 16:09:44 METDST 1997) Minor failures
2bit coding is not supported by vlbag DAR. I changed cables from
MKIII sampler to VLBA sampler and a line from the equip.ctl file
from vlbag to vlba at 08:29. Continous formatter problem.
Lost scans: (10:49 ->10:56, 16:45 ->17:19). Formatter jump of 3 secs.
at 11:32:40. Corrected after the scan.
Calibration (Fredrik T Rantakyro at Wed Nov 5 16:05:44 MET 1997) Minor failures
The ANTAB formatted TSYS information can be found on VLBEER in the
SEP97 directory. The file is called gp016.antab and contains the
latest antenna calibration information.