Sep 1997 (session 3)
To give feedback on this experiment, click
Medicina (M. Bondi at Tue Sep 30 23:14:23 METDST 1997) Success
Medicina (orfei at Wed Oct 1 11:00:08 METDST 1997) Minor failures
we realize that the patch panel was not changed respect the previous exp.
For the c97x3 it was LO1=1h,2h,3h,4h,5h,6h,7h while for this experiment is
LO1=1h,3h,5h,7h,9h,11h,13h. So only 1,3,5,7 videoconv. were properly patched
Effelsberg (dgraham at Wed Oct 1 12:47:38 METDST 1997) Minor failures
Lost 0900-0907 (not on tape). Phase cal off until 0920. Rest OK.
Onsala (John Conway at Fri Oct 3 11:16:14 METDST 1997) Minor failures
GPS-FMOUT monitoring of formaater GPS time difference shows
jump of 0.43msec in formatter around 1543 (and 1 microsecond jump
30 minutes before). The formatter clock was reset at around 16:30UT.
Note that on resetting the year was set to 97 not 1997, was noticed
and corrrected at end of experiment.
Calibration (Fredrik T Rantakyro at Wed Nov 5 16:08:01 MET 1997) Minor failures
The ANTAB formatted TSYS information can be found on VLBEER in the
SEP97 directory. The file is called el20a.antab and contains the
latest antenna calibration information.