Sep 1997 (session 3)
To give feedback on this experiment, click
Medicina (GMaccaferri at Wed Sep 10 16:04:57 METDST 1997) Success
New FS-9.3.15 used. maser resyncronized
Onsala (John Conway at Wed Sep 10 18:07:04 METDST 1997) Minor failures
Due to FS upgrade, had problems with FS to antenna computer
control at begining of expt, had to enter source coordinates
manually from antenna computer. One scan missed due to having
to recompile software. Problem wih phase-cal, expect no phase-cal
on tapes. In second half computer control restored, expect only
one scan lost.
Cambridge (Alastair Gunn at Thu Sep 11 10:09:32 METDST 1997) Minor failures
FS 9.3.15 upgrade prior to start. Tsys logging is not possible on the
Cm link. Error in MERLIN/VLBI switch led to dual pol. recorded until
13:19 UT. At single pol. BBCs 4,6,8 do not have signal due to
restrictions on Cm link. BBC03 USB effected by RFI from the 163 MHz LO.
Effelsberg (dgraham at Mon Sep 15 10:46:55 METDST 1997) Success
OK. FS patched to 9.3.15
Metsahovi (Kirsi Karlamaa \& Kaj Wiik at Tue Sep 16 12:56:20 METDST 1997) Success
Wet clouds, no rain, otherwise succesfull.
Noto (C.Trigilio at Mon Sep 22 16:23:37 METDST 1997) Success
FS Version 9.3.15 was installed before the start of the observation. Phase-cal used for the first time in Noto at 22GHz. No problem reported. Cloudy sky.
Calibration (Fredrik T Rantakyro at Wed Nov 5 15:40:57 MET 1997) Minor failures
The ANTAB formatted TSYS information can be found on VLBEER in the
SEP97 directory. The file is called c97k3.antab and contains the
latest antenna calibration information.
JIVE (Lorant Sjouwerman at Fri Dec 12 14:41:35 MET 1997) Success
Network monitoring report of K-band session