5 cm Session
Session: Sep 2000 (session 3)
Wavelength: 5 cm
To give feedback on this session, click
5 cm Session Feedback
Effelsberg (A. Kraus at Thu Sep 21 14:35:52 METDST 2000) Minor failures
Unfortunately, for this receiver both polarizations are exchanged.
We refrained from changing that during the session to avoid confusion.
Even worse, the RCP-channel of the receiver has a much higher system
temperature than the LCP-channel. And: Tsys varies strongly with
frequency. More details will follow.
Torun (KBorkowski at Mon Sep 25 11:52:22 METDST 2000) Minor failures
Tsys calibration in LCP channels is unstable or wrong altogether.
JIVE (Lorant Sjouwerman at Tue Oct 10 15:31:48 METDST 2000) Minor failures
In Socorro we found the polarizations for Torun to be swapped (like Eb)