Session: Oct 2014 (session 3)
Wavelength: 6 cm
To give feedback on this experiment, click
(evn at Tue Oct 21 14:05:37 2014) Success:
(evn at Tue Oct 21 14:07:41 2014) Success:
(evn at Tue Oct 21 14:08:10 2014) Success:
(Zhukova, Lysakova at Tue Oct 21 14:38:47 2014) Success:
Hartebeesthoek (Jonathan Quick at Tue Oct 21 18:44:56 2014) Minor failures:RECE
Receiver partially warmed up during experiment due to an electrical breaker failing and then re-cooled once power was restored to the helium compressor. No other known problems
(LUIGI PAPALEO at Tue Oct 21 20:02:52 2014) Success:
(Daniel Sastre Walter at Tue Oct 21 20:07:09 2014) Minor failures: RECO
Some problems to record properly from scan 38 to scan 43
(Shumilova at Tue Oct 21 20:42:48 2014) Success:
(Ptitcin A. at Tue Oct 21 20:43:17 2014) Success:
Onsala (Jun Yang at Wed Oct 22 06:26:06 2014) Minor failures:
The first 10 minutes were out due to a change of antenna gearbox. Proper Tsys data were not available until the noise source were reset to normal MK4 calibration mode. At 9:15 UT, there was an error: ERROR sc -24 setcl: Mark 5B syncerr_gt_3. CONSIDER using fmset 's' option to fix. I tried to run fmset for a few times, while, MK5B was busy and refused to sync time during the gap (no recording).
(Jimmy Cullen at Wed Oct 22 08:22:26 2014) Failure: WIND
Observation missed due to high winds
(Geert Kuper at Wed Oct 22 12:03:57 2014) Success:
(K.Borkowski at Thu Oct 23 11:27:07 2014) Success:
No known problems.
Onsala (Jun Yang at Mon Oct 27 06:59:03 2014) Minor failures:
Switching on and off noise source had a long time delay (~20 seconds) in the first hour and so there were no Tsys measurements. After that, the bug was found and fixed by our software engineer.
There was an error:
2014.294.09:15:50.22?ERROR sc -24 setcl: Mark 5B syncerr_gt_3, CONSIDER uisng fmset 's' option to fix.
According to its suggestions, we tried to run fmset to fix it during the gap for a few times, while MK5B reported too busy to service request.
(Uwe Bach at Tue Oct 28 13:04:00 2014) Minor failures: WIND
There were two stops due to high wind at