Session: Oct 2009 (session 3)
Wavelength: 0.7 cm
To give feedback on this experiment, click
Effelsberg (Uwe Bach at Thu Nov 12 10:41:04 2009) Success :
Of course ;-)
Onsala (Michael Lindqvist at Fri Nov 13 09:15:11 2009) Success :
Some rain first part. Otherwise no known problems.
Robledo (Esther Moll at Fri Nov 13 13:18:58 2009) Success :
1. Formatter was on sync.
2. No antenna problems.
3. Mark5 recorder#1 performance OK.
1000766386552 bytes of data recorded on module MPI-0067/1480/1024.
All scan_check correct.
4. LCP diode fired for every preob.
Generated antabfs file using antabfs application.
Sent to vlbeer archive observation log and antabfs file.
5. gps-fmout value included in the log at the beggining of the observation.
6.Performed pointing from 18:34UT to 18:51UT in DR21 (found EL=-4mdeg, XEL=-7mdeg offsets) and 3C345(found EL=-4mdeg , XEL=-3mdeg offsets). Applied EL=-4mdeg and XEL=-5mdeg pointing offsets for the rest of the run.
Noto (P. Cassaro at Mon Nov 30 10:12:09 2009) Minor failures : RECO
A couple of trouble periods with the MK5: from 15:00 to 15:50 UT and from 18:00 to 18:30 UT