Session: Oct 2008 (session 3)
Wavelength: 6 cm
To give feedback on this experiment, click
Medicina (Roda at Wed Oct 22 12:08:54 2008) Success :
Shanghai (Bo Xia at Wed Oct 22 12:14:47 2008) Success :
Torun (K. Borkowski at Wed Oct 22 12:24:57 2008) Minor failures : RECO TSYS
There are no cal data (noise diode disconnected by some accidental event) till 7:15 UTC. At 11:27 communication error occurred so probably no data written during the final 33 minutes. We will not use this diskpack (0S0D-066) although it is nearly half empty.
Urumqi (Wenjun Yang at Wed Oct 22 12:29:41 2008) Success :
Onsala (Michael Lindqvist at Wed Oct 22 14:43:54 2008) Failure :
To much wind. We also have mechanical problem with the antenna.
Effelsberg (Uwe Bach at Wed Oct 22 15:06:53 2008) Success :
Westerbork (Geert Kuper at Wed Oct 22 21:30:21 2008) Minor failures : FORM
Wb used in phased array mode.
Mark5B formatter gave error 'Mark5B sync error gt_3'
07:57 - 08:16 UT the DZB backend was rebooted.
Noto (P. Cassaro at Mon Nov 3 10:15:29 2008) Success :