Session: Nov 1998 (session 4)
Wavelength: 6 cm
To give feedback on this experiment, click
Medicina (Sergio Mariotti at Fri Nov 20 09:09:07 MET 1998) Success
Torun (K.Borkowski at Fri Nov 20 10:21:15 MET 1998) Minor failures
Scan 18:57-19:02 UT off source (antenna still slewing).
Calibration of Tsys was unreliable most of the time (some technical problem).
Effelsberg (dgraham at Fri Nov 20 14:58:17 MET 1998) Minor failures
snow stopped antenna 1335-1406UT
Noto (C.Trigilio at Mon Nov 23 10:19:28 MET 1998) Minor failures
Observations started with wrong configuration of Field-system. Wrong receiver LO (data lost) till 15:26.
Westerbork (Tony Foley at Tue Nov 24 16:23:20 MET 1998) Minor failures
Possibly bad polarization. From 1330 to 1515 sources were observed
using the wrong epoch .
Many of the scheduled scans were outside our Hour Angle range, and
so could not be observed
Jodrell (Alastair Gunn at Wed Nov 25 07:46:02 MET 1998) Minor failures
FS crash at start of schedule. Everything working by 15:05 UT.
Telescope encoder failure at 04:36 Ut, data lost until 04:42 UT.
Incorrect antenna.ctl file used, should be Mk2.
Calibration (Jean\-Francois Desmurs at Fri Dec 11 16:13:33 MET 1998) Success
The ANTAB formatted file for this experiment can be found on VLBEER
in the nov98 directory under the name:
ed009.antab. Read the comments in the file to get further