Session: May 1999 (session 2)
Wavelength: 6 cm
To give feedback on this experiment, click
Medicina (Morsiani Marco at Thu Jun 10 13:49:28 METDST 1999) Minor failures
Experiment started at day160 time 22:13:00
Torun (K. Borkowski at Thu Jun 10 14:16:24 METDST 1999) Minor failures
The BBC No 3 was noted to malfunction at 7:55 UTC and later at
irregular periods. Only 3 last minutes of the last scan useful
due to a long (about 300 deg) slew to the 3C84 source from previous
Effelsberg (A. Kraus at Thu Jun 10 14:57:43 METDST 1999) Minor failures
Observation done in VLBA-mode using the VLBA terminal. Scans between
UT 0600 - 0620 might be bad due to problems with the FS computer.
Noto (C.Trigilio at Thu Jun 10 15:12:08 METDST 1999) Minor failures
Only 8 minuts lost from 3:30 to 3:38 for antenna in prelimit.
Westerbork (Tony Foley at Fri Jun 11 13:44:07 METDST 1999) Minor failures
Most of the time had 12 telescopes in array (except from 05:39 to
05:52UT when it went to 11). Bad receiver removed about 07:30 UT
and replaced at 07:55UT.
Playback of this tape shows very pronounced "Once-Around" instability
in signal level on all passes (top reel motor effect).
Jodrell (Alastair Gunn at Fri Jun 18 10:55:56 METDST 1999) Success
No known problems.
Calibration (Jean\-Francois Desmurs at Mon Jun 21 16:11:05 METDST 1999) Success
The ANTAB formatted file for this experiment can be found on VLBEER
in the may99 directory under the name:
Read the comments in the file to get further information.