Session: Jun 2014 (session 2)
Wavelength: 6 cm
To give feedback on this experiment, click
(m.paterno at Tue Jun 10 09:21:09 2014) Minor failures: LATE
Hartebeesthoek (Jonathan Quick at Tue Jun 10 10:57:44 2014) Success:
No Known problems
(Ruben Fernández at Tue Jun 10 15:02:26 2014) Success:
Svetloe (Andreeva T. at Tue Jun 10 16:53:17 2014) Success:
Zelenchukskaya (Kiselev P. at Tue Jun 10 16:53:37 2014) Success:
Badary (Alakova, Panova at Tue Jun 10 16:53:54 2014) Success:
(Bo Xia at Tue Jun 10 23:57:46 2014) Success:
(Geert Kuper at Wed Jun 11 10:27:14 2014) Success:
(Jimmy Cullen at Wed Jun 11 14:10:59 2014) Minor failures: ANT
Cal diode hardware failure detected and repaired. No cal diode data before 09:31UT
(K. Borkowski at Thu Jun 12 07:26:58 2014) Minor failures:
Problems at disk-pack change (at 12:08 UTC) similar to those described in the feedback for EG078B. It required rebootting the recorder. Recording resumed at 12:42.
(Jun Yang at Thu Jun 12 09:10:34 2014) Success:
No known problems.
(Jimmy Cullen at Thu Jun 19 14:31:53 2014) Minor failures:
Potential LO problem at start of observation
(Uwe Bach at Mon Jun 23 13:47:11 2014) Success:
During the start 3-5 UT the antenna needs about 10s more time to switch between sources than SCHED predicted. Have to look if SCHED or antenna speed is wrong.