Session: Jun 2010 (session 2)
Wavelength: 6 cm
To give feedback on this experiment, click
(Morsiani at Sat May 29 10:26:47 2010) Success:
(Geert Kuper at Sat May 29 11:47:13 2010) Minor failures: LATE
12 Telescopes in tied array.
Missed the first 27 minutes due to TMS software problem.
(Wenjun Yang at Sat May 29 15:23:43 2010) Minor failures: FORM
problems with 1pps, connect 1pps to back of Mark5B+, use 'mk5=1pps_source=altB' to sync it. some data no useful.
Onsala (Michael Lindqvist at Sat May 29 20:41:37 2010) Minor failures:
Mark5 hanged around 00:10, see log for details. Had to reboot the Mark5. Bad module.
(Paco Colomer at Sun May 30 17:17:57 2010) Failure: ANT
The antenna cannot be moved. This experiment has not been observed.
(K. Borkowski at Mon May 31 11:13:10 2010) Minor failures: RECO
During last about 30 minutes there were problems with Mk5A. The now familiar warning "Recording throttled (can't keep up)" popped up and the four LEDs on the pack front went out.
Effelsberg (Uwe Bach at Wed Jun 02 08:02:41 2010) Success: