Session: Jun 2010 (session 2)
Wavelength: 6 cm
To give feedback on this experiment, click
(Geert Kuper at Fri May 28 19:44:04 2010) Success:
12 telescopes in array.
No known problems.
(roma at Fri May 28 21:30:06 2010) Success: RFI
Success. All scan recorded. Some problems for severe RFI.
(Wenjun Yang at Sat May 29 15:14:37 2010) Minor failures: FORM
problems with 1pps come from mark5B sampler, it can not sync Mark5B+. use 'fmset' or 'gpstime' manually to fix it, parts of data perhaps no useful.
Onsala (Michael Lindqvist at Sat May 29 20:38:56 2010) Minor failures:
Some tracking problem around 12:52 UTC. See log.
(Paco Colomer at Sun May 30 17:14:31 2010) Failure: ANT
The antenna cannot be moved. This experiment has not been observed.
(K. Borkowski at Mon May 31 10:58:16 2010) Minor failures: RECO
About an hour after start could not be recorded. Mk5 said "Warning: Recording throttled (can't keep up)". Finally hard reset helped.
Effelsberg (Uwe Bach at Wed Jun 02 07:36:46 2010) Success:
Missed one scan 13:15 because of long way in azimuth.