Session: Jun 2009 (session 1)
Wavelength: 3.6 cm
To give feedback on this experiment, click
Onsala (JFD at Wed Jun 17 02:53:03 2009) Success :
OK, every things went fine.
Shanghai (Bo Xia at Wed Jun 17 20:57:08 2009) Success :
Urumqi (Wenjun Yang at Thu Jun 18 01:27:43 2009) Success : HIGH
Onsala (Michael Lindqvist at Thu Jun 18 06:36:41 2009) Success :
No known problems.
Medicina (Andrea Orlati at Thu Jun 18 08:39:48 2009) Minor failures : RECO
Lost scans at the bigining due to MK5 module failure. Module replaced with a new one (last available in stock) but not enough large to fit the whole experiment.
Yebes (JFD at Thu Jun 18 09:05:41 2009) Failure :
Yebes Update: It seems that our maser failed during this observations. Data are probably corrupted and useless.
Effelsberg (Uwe Bach at Thu Jun 18 10:16:28 2009) Success :
Westerbork (Geert Kuper at Thu Jun 18 15:30:14 2009) Success :
Wb used in phased array mode.
No RadioTelescope B,C and D in the adder.
Because this was a non standard setup the TADU setup was done by hand. It is also possible that this does not exactly match the vexfile. See the remarks and Mark Kettenis email about N09SX1 for the wsrt.
Video settings per Band:
BAND-0 BAND-1 BAND-2 BAND-3 BAND-4 BAND-5 BAND-6 BAND-7
8360.990 8360.990 8376.990 8376.990 8392.990 8392.990 8408.990 8408.990
TOM0 n n n n 1R 3R 5R 7R
TOM1 n n n n 0R 2R 4R 6R
Mark5B mode 0xFFFFFFFF, but data should be in 0xFF00FF00
Only X-band is recorded.
Noto (P. Cassaro at Thu Jun 25 16:19:56 2009) Minor failures :
bbc#11 unlocked for the whole experiment
Yebes (Paco Colomer at Thu Jul 2 09:47:06 2009) Failure :