Session: Feb 1999 (session 1)
Wavelength: 6 cm
To give feedback on this experiment, click
Medicina (Andrea Maccaferri at Mon Feb 22 02:14:05 MET 1999) Success
Effelsberg (dgraham at Mon Feb 22 12:35:05 MET 1999) Failure
One of the azimuth gearboxes is kaputt. No observation.
Noto (C.Trigilio at Mon Feb 22 23:39:23 MET 1999) Minor failures
Lost first 3 passes (from 17:00 to 18:12) for a Field system problem.
Lost last pass (from 23L:00 to 23:24 in tape #2) for a problem in the recorder. No data in the second tape; only the first tape will be sent.
Onsala (A. Polatidis at Tue Feb 23 18:51:46 MET 1999) Minor failures
Lost data for about 30 min before 2144, due to problems with antenna tracking. Due to a failure in the cooling system the maser failed around 0257 and the system was not phase locked after that time.
Calibration (Jean\-Francois Desmurs at Thu Feb 25 12:45:33 MET 1999) Success
The ANTAB formatted file for this experiment can be found on VLBEER
in the feb99 directory under the name:
gb029.antab. Read the comments in the file to get further
information. (Rem : No Tsys values for On in RCP)