Session: Feb 1997 (session 1)
Frequency: 6 cm
Obs time (UT): EU 19:30(21/02)-07:00(22/02)
EU Mk3 Tapes/St.: 2
To give feedback on this experiment, click here
Medicina (A. Cattani at Sat Feb 22 02:24:40 EST 1997) Success
Onsala (Antonis Polatidis at Sun Feb 23 09:04:28 EST 1997) Minor failures
Due to high winds the antenna was stowed until UT 0247. No known problems with the recorded data.
Effelsberg (dgraham at Mon Feb 24 03:29:45 EST 1997) Failure
Telescope drive failed from 0000 until end of experiment
Westerbork (Tony Foley at Mon Feb 24 10:54:14 EST 1997) Minor failures
One restart (maybe phase jump) at 06:30 UT
HartRAO (Jonathan Quick at Mon Feb 24 21:14:12 EST 1997) Minor failures
Read heads failing, write head status unknown
Shanghai (Huang Xinyong at Wed Mar 5 04:57:52 EST 1997) Minor failures
Clock offset = 0.10 msec. Tape vacuum failed between 19:35:10
- 19:37:04. Weather fine.
Jodrell (Alastair Gunn at Wed Mar 5 09:06:00 EST 1997) Minor failures
FS re-started (new log) due to problems with vacuum at start of
experiment. Data should be good from 20:41 (number 5) scan. FS
crashed and re-started at 02:16 UT. No other known problems.
Tapes EVN00043,B284 and MPIR1221,F37B sent to Bonn via Woodcroft
Cambridge (Alastair Gunn at Wed Mar 5 09:10:15 EST 1997) Success
No known problems. Tapes NAO00004,B6F3 and NGS00390,32EF sent
to Bonn via Woodcroft on 24/02/97.
Calibration (Fredrik T Rantakyro at Thu May 8 11:35:22 METDST 1997) Success
The ANTAB file for the experiment has been placed on VLBEER in the
Feb97 directory with the name eh1.antab. This file contains the
preliminary calibration results and may be updated when I receive
updated information from the telescopes and the results from the
datareduction of the calibration experiments.
MPIfR (Scott Aaron at Fri Mar 20 10:55:37 MET 1998) Minor failures
Correlation completed. Usual tracking problems with Torun.
Frequent problems with the Jodrell tapes too. No other problems