Title: CSS/GPS Sc.
Session: Feb 1997 (session 1)
Frequency: 6 cm
Obs time (UT): EU 03:00(16/02)-19:30(16/02)
EU Mk3 Tapes/St.: 4
To give feedback on this experiment, click here
Medicina (Franco Mantovani at Sun Feb 16 15:06:19 EST 1997) Success
Onsala ( at Sun Feb 16 16:27:50 EST 1997) Success
No known problems
Effelsberg (dgraham at Mon Feb 17 02:52:23 EST 1997) Minor failures
Formatter probably 1 sec slow. Used caltemp=10 instead of 1.4K so
Tsys calculation wrong.
Westerbork (Tony Foley at Tue Feb 18 10:06:48 EST 1997) Success
Please note that we needed to make lots of hand edits
to the array schedule to allow for gaps where sources
were out of Hour Angle range.
Torun (Roman Feiler at Wed Feb 19 18:11:01 EST 1997) Minor failures
4:32-4:44, 6:14-6:34, 19:16-19:30 off source due to power failures
Jodrell (Alastair Gunn at Fri Feb 21 03:31:57 EST 1997) Failure
POLARISATIONS ARE CROSSED. Therefore no useful data.
Cambridge (Alastair Gunn at Fri Feb 21 05:20:27 EST 1997) Minor failures
No major problems. FS crashed at about 14:51 so scan 24 missed.
Weather cloudy, windy. Tapes EVN00120,0AAD EVN00101,6A9D and
HS004353,EE4F sent to Bonn via Woodcroft on 17/02/97.
Shanghai (Huang Xinyong at Wed Mar 5 04:45:17 EST 1997) Success
Clock offset = -0.06 msec. No problems, weather fine.
Calibration (Fredrik T Rantakyro at Thu May 8 11:26:20 METDST 1997) Success
The ANTAB file for the experiment has been placed on VLBEER in the
Feb97 directory with the name ef2b.antab
MPIfR (Scott Aaron at Tue Sep 23 14:41:56 METDST 1997) Minor failures
Shanghai 100% data loss--telescope not connected to recorder.
MPIfR (Scott Aaron at Wed Dec 17 09:49:55 MET 1997) Minor failures
Correlation complete. No fringes to Shanghai. Jodrell Bank swapped
polarizations, so no data. Tracking problems on the Torun tapes, so
all baselines to Torun deleted from recorrelation list. Everything