Session: Feb 2017 (session 1)
Wavelengths: 18 21 18 6 90 1.3 0.7 1.3 0.7 cm
To give feedback on this entire session, click
Entire Session Feedback
(Mikhailov A. at Fri Mar 10 10:17:26 2017) Minor failures: RECO
Due to Mark5 disk pack problems data were lost from experiments
F17L1, EP103A, EM127A
(P.Cassaro at Sat Mar 11 10:57:34 2017) Failure:RECE
The receiver show problems and the Q-band session cannot be observed.
(Uwe at Mon Mar 13 09:48:25 2017) Success:
LCP channel at Q-band is broken. All Q-band observations will be RCP only.
(Jun Yang at Wed Mar 15 16:22:22 2017) Success:
At Onsala, Chopper-wheel calibration is used to measure Tsys just before and after a
scan at 22 and 43 GHz. While, moving the chopper wheel in and out requests relatively
longer time than firing electronic noise diode. At the scan beginning, a few second data
might be affected because of the Tsys measurement and should be flagged out during
the data reduction.
(Cristina Garcia Miro at Tue Mar 28 12:56:15 2017) Success:
EVN session I 2017 supported with DVP s/w version RSO-6726-OP-A V1.0.1: 16bytes VDIF headers, 32 real USB/LSB channels recorded, SDK 9.2 and SScard f/w 16.45