Session: Feb 2002 (session 1)
Wavelength: 5.0 cm
To give feedback on this experiment, click
Effelsberg (A. Kraus at Wed Feb 6 08:31:50 MET 2002) Success
Torun (K.Borkowski at Wed Feb 6 11:58:28 MET 2002) Minor failures
No automatic Tsys calibration. Lost observations 9:40 to 9:58 because
of a false alarm (no signal on head caused by schedule-set wrong tracks was attributed [aptly in our case] by operator to recorder failure; thus terminal has been shut down and ... you know what).
Medicina (G.Maccaferri at Wed Feb 6 17:14:25 MET 2002) Success
Onsala (John Conway at Fri Feb 8 08:24:04 MET 2002) Failure
WE did not observe due to high winds. No tape
has been shipped
Jodrell_Bank (Alastair Gunn at Mon Feb 11 11:26:26 MET 2002) Minor failures
Tsys diode was accidentally switched off between about 10:20 and
13:20 UT - bad Tsys. Several scans on J2022+61 missed between 11
and 13 UT due to inadequate slew times. No other known problems.
Cambridge (Alastair Gunn at Mon Feb 11 11:26:55 MET 2002) Minor failures
Several scans missed due to inadequate slew times or cable
wrap problems. No other known problems.